Payment & Shipping Terms:
|Cooling Mode:||Water Cooled||Heater:||Iron-chrome Wire Heater|
|Interior Material:||Stainless Steel(SUS04)||Interior Size:||W800 Mm*H900 Mm*D700 Mm|
|Controller:||Touch Screen Programmable Controller||Temp Range:||-65℃ ～ +100℃|
1000Lwater cooled Rapid temperature change chamber with cable port Lighting device
The Rapid-Rate Thermal Cycle Chamber is ideal for test requiring quick changes 15°C/min ramp rate in
specimen temperature and 23°C/min ramp rate for air temperature. It covers various applications from
JEDEC, IEC standards to screening.
It features new technologies, such as a specimen temperature control function maintaining linear
specimen temperature change rates during rapid thermal cycling, and temperature ramp control.
Rational Construction and fast cooling rate
Frequency conversion control of wind speed.
Adopts international brand of imported parts for cooling system to improve operation reliability
Touch screen controller, friendly Man-Machine interface and safety device to ensure simple operation
and easy maintenance.
Wide variety of refrigeration available for performance and temperature range flexibility.
Programmable LCD touchscreen controller for product or air-change available.
|2.1 Volume||About 1000L|
|2.2 Interior size||
W800 mm*H900 mm*D700 mm
|2.3 Exterior size||
W1100 mm*H2080 mm*D2150 mm(Excluding the protruding part of the machine!)
Tips: For external dimensions, please confirm the three views according to the final design!
|2.4 Floor area||About 2.73m2；(confirm after signing the contract)|
3. The main technical parameters
|3.1 Test Conditions||
Cool method: water-cooled
Measured at water temperature of +25 ℃ under no load, the temperature and humidity performance test is measured according to the relevant regulations according to IEC 60068 -3; the sensor is placed at the air outlet of the air handling unit
|3.2 Temperature range||-65℃ ～ +100℃|
|3.6Temperature change rate||
Heat up rate:
-60℃～+70℃, full range average 10℃/min；(Load 30kg PCB module, no heat)
+70℃～-60℃, full range average 10℃/min；(Load 30kg PCB module, no heat)
|* Explanation on the average non-linear cooling and measurement curve of the whole process! (According to the acceptance standard of GB / T2423-5 or IEC60068-3-5)|
|3.7 Load||Load 30kg PCB module, no heat|
|3.8Noise||≦75 (dB) (The noise detection device is measured 1m away from the door of the chamber)|
|3.9 Meet the test standard||
GB/T 2423.1-2008(IEC68-2-1)Test A:Low temperature test method.
GB/T 2423.2-2008(IEC68-2-2)Test B: High temperature test method.
GJB360.8-2009(MIL-STD.202F)High temperature life test.
GJBl50.3-2009(MIL-STD-810D)High temperature test method.
GJBl50.4-2009(MIL-STD-810D)Low temperature test method.
We use PP Film wrapping the test chamber to protect the test chamber from moisture or being scratched
first, then use strong plywood case for easyload and transport, and to keep the chambers from potential
crash or impact.
If you face any problems in future during testing process we promise we will give you a reply within 48
hours and solution provide within 3 working days. We can offer operation video and English operation
muanual. We can provide video-conference. If customers require on-site service, the transport and travel
expense shall be borne by customer.
Contact Person: Annie Peng