Payment & Shipping Terms:
|Temperature Range:||+2℃ ～ +8℃||Temp. Deviation:||≦±2.0℃|
|Workspace Volume:||W 800 × H 1425 × D 840 Mm||Exterior Size:||W 1000 × H 1705 × D 1860 Mm|
high temperature test chamber,
low temperature test chamber
High And Low Temperature Environmental Control Chamber Single Door 1000L
Able to accurately simulate a wide range of complicated natural environments, and is suitable for
reliability test in industrial products. Meet GB5188.8.131.52.6-95 standard requirements of environmental
testing equipment and test methods for the basic parameters of electric and electronic products under
the condition of humidity, low temperature, high temperature, and constant heat.
Applicable to environmental adaptability and reliability test in such industrial units as electronics, electrical
appliance, battery, plastics, food, paper product, vehicle, metal, chemistry, building material, research
institution, inspection and quarantine bureau, university etc..
IV. Main Technical Parameters
(Performance: Air-cooled, Room temperature +25°C, no load)
|Temperature range||+2℃ ～ +8℃|
|Temperature performance measurement comply with related regulation of IEC60068-3 standard; Sensors placed in the air outlet.|
Overall structure and chamber was composed of three parts as below.
Insulation box, separate refrigeration units, and electrical control cabinet.
Workspace volume: W 800 × H 1425 × D 840 mm
Exterior size: W 1000 × H 1705 × D 1860 mm (not including the machine protruding part)
wall material: high-quality carbon steel with static color spray
inner wall material: SUS304 # matte stainless steel plate
Insulation materials: rigid polyurethane foam insulation layer + glass fiber.
|Door||Single door, left open,Heating wire was installed at the door frames to prevent condensation at low temperatures.|
|Observation window||With 1 set 460 × 560mm (width × height) observation window, multi-hollow electric insulation coated glass prevent condensation effectively|
|Lighting device||LED Lighting device* 1 located on observation window.|
Quality nickel-chromium alloy wire electric heaters,
Non-contact control mode (SRR).
|Water outlet hole||Equipped with studio condensate and unit condensate drain holes.|
|Cable port||Φ50mm*2 located on both sides (each*1) with rubber stopper and plastic cover|
|Shelf||Stainless steel frame *1, Exterior size:W730*H925*D760mm, There are three layers, of which the first layer is 165mm from the ground, the second layer is 380mm away from the first layer, and the third and second layers are 380mm apart.|
|Power distribution control cabinet||Total power circuit breaker, over-temperature protection.|
|Recorder||2-channel recorder * 1 (Brand: ternary).|
|UPS power||UPS power *1 ( Controllers and recorders :30min).|
|Chamber Insulation||The temperature is maintained between +2 ° C ～ +8 ° C within 15 minutes after the power failure.|
Environmental test chamber Standard
GJB150.3 high-temperature testing
GJB150.4 low-temperature testing
GJB150.9 humid testing
GB11158 Technical conditions for high-temperature testing box
GB10589-89 Technical conditions for low-temperature testing box
GB10592-89 Technical conditions for high and low temperature testing box
B/T10586-89 technical conditions for humidity testing chamber
GB/T2423.1-2001 Low-temperature testing methods
GB/T2423.2-2001 High-temperature testing methods
B/T2423.3-93 Testing Method for humidity Test Chamber
GB/T2423.4-93 Alternating hot and humid testing method
GB/T2423.22-2001temperature testing method
IEC60068-2-1.1990 low-temperature testing method
IEC60068-2-2.1974 high-temperature testing method
We use PP Film wrapping the test chamber to protect the test chamber from moisture or being scratched
first, then use strong plywood case for easyload and transport, and to keep the chambers from potential
crash or impact.
If you face any problems in future during testing process we promise we will give you a reply within 48
hours and solution provide within 3 working days. We can offer operation video and English operation
muanual. We can provide video-conference. If customers require on-site service, the transport and travel
expense shall be borne by customer.
Contact Person: Annie Peng