Difference between the real humidity and the target humidity of Constant temperature and humidity chamber
In the case of hot and humid test, the actual humidity will reach 100% or the actual humidity is very different from the target humidity. or the humidity value is much lower than target humidity.
There are two main reasons for this:
1. Wet cloth on wet bulb sensor drying caused.
Check the wet bulb sensor of water tank if water shortage, the water level in the tank is controlled automatic by a water level controller, check the water level controller water supply system is normal, water level controller whether work is normal.
2. Another may be caused by long time use of wet cloth, or water quality purity reasons, will make the wet cloth hard, so that wet cloth dry and can not absorb water.
What you need do is replacement or cleaning wet cloth.
The latter phenomenon is mainly the constant temperature and humidity chamber humidification system does not work, check the humidification system of water supply system, whether there is a certain amount of water to control the humidifier boiler water level control is normal, humidification boiler water level is normal.
If everything is normal, it is necessary to check the electrical control system, which requires professional maintenance personnel to repair. KOMEG, professional manufacturer will provide you professional services!