Payment & Shipping Terms:
|Temp Range:||-40℃ ～ +180℃||Temp Deviation:||≦ ± 0.5℃|
|Pressure Range:||Normal Pressure ～1k Pa||Heating Uptime:||0.7℃～1.0℃/minutes(Average )|
high temperature test chamber,
low temperature test chamber
Stainless Steel High And Low Temperature Test Chamber Two Sides Door Open For Oil Electronic Pump
Main technical parameters ( water-cooled, temperature +25℃, no load)
-40℃ ～ +180℃
5.Heating up rate&Cooling down rate
from 25℃ to +150℃ whole process average≧3℃/min Load 50 kg aluminum alloy
from +25℃ to -40℃whole process average ≧2℃/min Load 50kg aluminum alloy
The temperature performance test is based on the relevant provisions of the IEC 60068 -3
standard; the sensor is placed at the unit air outlet.
The test chamber adopts the overall structure, the upper part is the working chamber, the rear side is
the refrigeration unit, the lower right space (≧700mm) is vacant to install the hydraulic part device, the
right is the electric control cabinet, the control panel is located on the right front side; the width direction
opens the door , the right side is the air duct system; the air supply motor, the centrifugal fan blade, the
heater, and the evaporator.
Inner size: W 1800 × H 1200 × D 1200 mm
Outer size :W 2935 × H 2170 × D 1450 mm approx
Outer wall material: cold-rolled steel plate electrostatic double-sided spray, color ivory white;
Inner wall material: SUS304# matte stainless steel plate, thickness 1.5mm
Insulation material: rigid polyurethane foam insulation layer + glass fiber.
bottom bearing capacity: ≧150kg
The bottom of the workroom has a threaded port for mounting the work shelf
Standards Implement and Met
● GB-2423.1-2008(IEC68-2-1)Test A: Low Temperature Test.
● GB-2423.2-2008(IEC68-2-2)Test B: High Temperature Test
● GJB360.8-2009(MIL-STD.202F) High Temperature Life Test
● GJBl50.3-2009(MIL-STD-810D)High Temperature Test
● GJBl50.4-2009(MIL-STD-810D) Low Temperature Test
● GB2423.3-2008 (IEC68-2-3) Test Ca: Constant Hot and Humid Test
● GB2423.4-2008 (IEC68-2-30) test Db: Alternating Hot and Humid Test
● GJBl50.9-2009 (MIL-STD-810D) Damp heat test method
1. Are your company a trading one or a factory?
Factory ,28 years focusing on test instruments field,4 years exporting experience.
2. After an order is placed, when to deliver?
Generally about 20 days, If we have finished products, we can arrange shipment within 3 days.
Please be aware that our production lead times depend on specific items and item quantities.
3. What about the warranty with after - sales services?
12 months warranty.
After the warranty, the professional after-sale service team will provide perfect technical support,and help customer solve problems encountered while using our product, handles guest problems and complaints in a professional and timely.
4. What about the services and product' quality?
Each instruments must be carried out 100% quality examination and test, the finished products must through a third party calibration institutions before shipping and delivery goods.
If you need more information about Temperature Chamber Calibration/ Temperature Humidity Test Chamber, please don't hesitate to send your enquiry, we will support more information for your reference.
Contact Person: Anna Hu