Payment & Shipping Terms:
|Workspace Volume (W X H X D):||W1000 X H1000 X D1000mm||Exterior Size (W X H X D):||W1280 X H2160x D1950mm|
|Temperature Range:||-70 °C To 150 °C||Material:||SUS304 Stainless Steel|
|Controller:||LED Touchscreen 10 Inch|
high temperature test chamber,
low temperature test chamber,
Temperature Test Chamber With Safety Lock
Stability High And Low Temperature Test Chamber With Safety Lock , Environmental Protection
1. Explore High-low temperature test chambers:
High-low temperature test chamber is special-purpose thermostatic and humidistatic equipment. It is able to accurately simulate the climate conditions in different environments and is applicable to research institutes, university laboratories and production sectors in sanitary and anti-epidemic, environmental protection, forestry and animal husbandry industries for culture preservation, plant cultivation and breeding experiments of molds, bacteria, and microbes.
Protection function: leakage protection, over temperature limit alarm, door open alarm, water level alarm etc.
Automatic fault detection and display.
3. Performance description
Microprocessor fuzzy PID control, programmable multi-segment parameter setting: 30 segments 99 cycles design.
Operation system based on WINDOWS 2000 system with large-capacity memory and High frequency CPU.
90 days long memory storage.
C.H.heating system: improve the use of excess heat to reduce heating time.
C-changer defrosting technology: automatically adjust refrigeration output according to different environment, combined with the C.H. heating system to achieve automatic quickly defrosting effect.
USB interface and RS485 interface are standard configuration.
Optional: printer (can print into curve);
One choice from two: GPRS SMS Alarm, computer monitoring.
4. Technical Parameter:
|1000L chamber volume|
|Temp. range||-70°C ～150°C|
|Temperature stability||High±0.5°C ,low ±1.0°C|
|Heat up time||-70°C ～+100°C,within 60 min|
|Pull down time||+20°C ～-70°C,within 80 min|
|Interior Dimension(mm)||W1000 x H1000 x D1000mm|
|Exterior Dimension(mm)||W1280 x H2160x D1950mm|
|Interior material||Stainless Steel Plate ( SUS 304)|
|Exterior material||Baked Painting Steel Or Stainless steel ( SUS304)|
|Insulation material||Rigid polyurethane Foam and Glass Fiber Wool|
|Total power||18KW ( heater 6KW included)|
|Ambient Temp.||+5°C ～+35°C|
|Power Supply||AC 380V±10％ 50Hz ,3 phase 4 wires +Ground Wires|
|Sight window size||460 X 560mm|
|Sample holder||2 pcs|
|Fittings||cable port(portΦ50mm)*1, chamber lamp*1, power cable *2m|
|Safety device (standard)||
No fuse breaker,over pressure,over heat&over current protect,over temp.
protect, over load protect for blower, dry heat protect
|Control System||Balanced Temperature & Humidity Control (BTHC)System|
|Refrigeration system||Air Cooled (Water Cooled is option)|
Single stage refrigeration system,
|Controller||7-inch LCD Touch Screen panel|
|Operation model :Program or fix point running|
|Memory capacity :120programs,1200steps,|
|All repeat 999 cycles, part repeat 99 cycles|
Note: 1. We reserve the right to change specifications without prior notice 2. Customized sizes and configurations available
5. High temperature limits of test chambers
There are many types of test chambers, but they all high and low temperature limits. Let's take a look at the typical high temperatures thresholds and why they exist.
Test chambers usually have wire heaters made of a material called Nichrome. By placing successive loops of heater wire, a greater amount of heat that is generated. The result looks a lot like the inside of a toaster. Fans circulate the air past the heaters, so the samples in the chamber are not exposed to radiant heat. It is very easy to add heaters and increase the speed and / or high limit of a chamber.
KOMEG chambers typically have a high temperature limits under 200°C. Here is a break-down of why a chamber's high temperature limit exists:
150°C: There are two reasons we may limit a chamber to 150°C. First, there may be a limitation of the materials used inside the chamber. Secondly, for extremely fast chambers, we may limit the high temperature to avoid problems with accidentally tripping overheat protection devices.
170-180°C: Most models stop at this temperature range. There is a risk of burning the oil inside the refrigeration's cooling coil.
Contact Person: KOMEG